AED Provides operating capability for basic users.
Manual Provides operating capability for advanced users.
ECG Provides ECG display capability with 3-wire ECG cable.
Setup Allows user to configure the device.
Data Transfer Allows user to transfer patient data.
Auto Test Provides daily automatic tests of hardware and software.
Controls: On/Off, Shock, Menu, Two (2) configurable soft keys.
User Defined Options:
Device ID Assigns unique identifier to particular device.
Energy Sequence User configurable from 150 to 360 joules.
Flexible Energy Increases only after a lower energy was unsuccessful.
Auto Analyze User can configure device to auto analyze, auto analyze after
first shock, or prompt user to push analyze key before each analysis period.
CPR Time (Post shock or after no shock advised) User configurable
15, 30, 45, 60, 90, 120, 180 seconds.
Voice Prompt Volume Allows user to change speaker volume.
ECG Display (If option purchased) Turns display on/off for AED mode.
Motion Detection User defined On/Off (default On).
Service Alert Audio alarm if the device needs servicing. Configurable on/off.
Manual Access (If ECG display option purchased) Devices configured with an ECG
display may be set up to allow user to initiate a charge and shock without analysis.
cprMAX Technology Settings:
Initial CPR User defined time for CPR after first analysis regardless of analysis decision.
Can be set to OFF, 15, 30, 45, 60, 90, 120 and 180 seconds.
Pre-shock CPR Allows for CPR while device is charging. Can be set to OFF, 15,
or 30 seconds.
Stacked Shocks (ON/OFF) When Off, allows for provision of CPR after each shock.
Pulse Check (Always, After Every NSA, After Second NSA, Never) Allows device to
prompt for a pulse check either after each shock, after every NSA pulse check, or never
prompt for a pulse check (default Never).